Short Term Run Plan
Do reload this page, it gets updated frequently.
- 1 Current Experiment
- 2 Notes
- 3 Run conditions
- 4 Run plan
- 5 Shift duties
- 6 Special instructions
Rare Eta Decay & GlueX II (with DIRC)
- Experiment number: E12-12-002 (use in BTA shift information)
- Spokesperson: C. A. Meyer
- All GlueX detectors (including DIRC)
Hall D counting room whiteboard
- Radiator: JD70-103 50 um diamond (4 orientations: 0, 90, 45, -45)
- Collimator: 5.0 mm
- TPOL converter: Be 75 um
- Runs are no more than 2 hours (~400M triggers)- see whiteboard for the production cycle
- Target: Liquid Hydrogen
- Setup: FCAL_BCAL_PS_m9
- Production: PHYSICS_DIRC
- (Raw mode: PHYSICS_DIRC_RAW)
See the daily plan in Run Coordination Meetings:Spring2023 and also the counting house whiteboard
It's good practice to refer to our hall as Delta when talking to MCC, to make sure that there is no acoustic confusion between us and Hall B.
"Alt L" brings up the screenshot tool for making E-log entries
FCAL Base lockup
If FCAL HV bases lose communication and stop providing high voltage, refer to https://halldweb.jlab.org/hdops/wiki/index.php/Forward_Calorimeter_Shift#FCAL_Base_lockup
CDC Weather alarm
aka Alarm on HD:coda:cdc:p_diff. See https://logbooks.jlab.org/entry/4117725
- Stop run - Acknowledge alarm and document it - Start new run
We are temporarily using Hall B sensor for our alarm, see Alternative Barometric Pressure Measurement
DIRC Communication Error at Run Start
When starting a new run, one of the DIRC FPGA boards may lose communication resulting in a Magenta communication error in the Alarm Tree, refer to this page for a fix.
After beam restoration
- See Beam_Line_Detectors_Shift#Collimator_and_Radiators on how to move radiators and collimator
- check the radiation levels without radiator and then with the radiator
- see the whiteboard (bottom RHS corner of the RH board) - there are columns of values expected without and with the radiator
- retract the radiator
- collimator in blocking position
- ask MCC to do their harp scan
- our harp scans: follow the instructions in Beam_Line_Detectors_Shift#Radiator_Harp_Scan
If you have to restart rootspy, Online_Monitoring_Shift#Routine_Operation, and it crashes, you might have to restart the monitoring process, find help at Online_Monitoring_Shift#Starting_and_stopping_the_Monitoring_System.
Call the expert. In case they suggest that you restart the DAQ, find information at Data_Acquisition_Shift.
- The BTA guidelines are at BTA_guidelines.
After loading the EPICS values, you may change the values, eg round them up or down.
- BANU values should be reflected in the PCC (or UED) value (else the aftermath doesnt work out for RC), put a comment for > 5minutes.
- UED: unplanned experiment down (hardware failure).
Bottom Alcohol Bubbler
- We are not using the bottom alcohol bubbler. It is expected that its level is "ADD" and that it is not bubbling log entry