Run Coordination Meetings: Fall 2014
Daily Meetings, 8:45am, Hall D conference room. Held by the Run Coordinator. Present Run Coordinator: Alexandre Deur
To connect from the outside by phone please do the following
2.) To join via Polycom room system go to the IP Address: 220.127.116.11 (bjn.vc) and enter the meeting ID: 660743227.
3.) To join via phone, use one of the following numbers and the Conference ID: 660743227.
4.) The moderator code is 8394
Dial Either +1 408 740 7256 or +1 888 240 2560 for US or Canada
Specific instructions for connecting from JLab's Hall D conference room:
- Turn polycon on if necessary (do it before turning the computer on)
- With the polycon, place a call at 18.104.22.168
- Press # to enable the polycom keypad, then enter the meeting id: 660743227 and #
- You may have to unmute the microphone: #*4
- Turn the computer on if needed
Meeting General Agenda
Sept. 30th 2014 meeting
- Report relevant updates from 7:50 and 8:00 daily MCC meetings:
- Accelerator schedule:
- Accelerator is aiming to start beam restoration on Oct. 3rd and start running 5.5 pass/10 GeV on Oct. 7th. This means locking the Hall around Oct. 10th.
- Accelerator schedule:
- Report on Work in the Hall
- Hall D is now a radiation aread. OSL need to be worn wherever you go in Hall D.
- Elton will conduct a SAF113 training on Friday early afternoon.
- Report on yesterday Hall D related events
- Hot Check OUT Status
- Installation punch list (items in red: must by Oct 6th ; Blue: Done by 2nd Fall run; Green: Nice to have it
The wire is installed. Tom measured the wire diameter without insulation to be 200 microns.
Chamber is closed and under vacuum, leak-checked and ready to go.
Pair spectrometer detector - Positioned on the stand. Survey is done.
- Profile monitors -
The two profilers are now installed. The upstream profiler (collimator cave) is reconnected.
- Upstream profiler has been uncabled and moved to its pre-solenoid position for survey. Hovanes can read scalers from it. Downstream profiler is cabled. Hovanes useed a source to identify the left/right and up/down signals. The wiring is correct.
- Active collimator -
Cables are terminated and in the Hall. All hardware work is completed.
- Trent will label the signals cable today and Alex B. will connect them and test the signal readout.
The light leaks of the SC are largely fixed. SC is mounted back.
- Tagger shielding - We should have the wall (final labyrinth configuration) mounted by Oct 10th.
Pair Spectrometer Magnet remote control - Control is now working. Hysteresis procedure will be done by hand.
- Pair Spectrometer Magnet Field readout Hall probe is here and will be installed by Wes. Wes is working on the gui.
Pair Spectrometer detector: finishing today.
ProfilersDownstream profiler when in front of Solenoid: today or tomorrow
Start counter: Survey should be done by the end of the day.
FCal: Still working on it. Should be completed today
CDC: Should be completed (confirmation?).
- Microscope: tomorrow.
- Beam pipe section, after the pair spec.: today
- Vacuum - On schedule. The last piece of beam pipe (the one after the Pair Spec.) to be put under vacuum today.
Amorphous Radiators - FSD done and tested.
- Halo Counters - All are installed and connected, except the ones in front of solenoid.
- Microscope - Microscope is in its correct position. Being cabled. Making an air outlet.
Tagger crates - Can now been readout.Will be removed on the 7th.
- Hodoscope checks - The few hodoscope segments that have been removed to install the microscope were placed back. Nathan is testing the detectors and electronics with DAQ. 131 (out of ~200) counters have been checked directly with a scope. Nathan will continue to check the rest of them. 7 out of 131 counters have problems (1 not responsive, 6 with cabling problems) and need to be fixed.
- Radiation monitors - The gamma probe in the collimator cave is installed. Readout gui to be made. The 2 gamma and the 1 neutron probes in the Tagger Hall will be installed today. Ion Chambers installed but one is not at the right location. This is being fixed.
- 2nd sweep magnet. (2nd collimator aperture) - not required for Fall run.
- Specific safety notice (from Tom/Tim/Mark):